Designed to aid in the testing of fiber optic cable when using an OTDR. The OTDR Launch Fiber box is used with Optical Time Domain Reflectometers (OTDR's) to help minimize the effects of the OTDR's launch pulse on measurement uncertainty. Available in many different configurations and fiber lengths.
-Pulse Suppressor, Launch Box, Deley Line, Installation/Testing, Training, Calibration
-Compound Latch for positive seal and easy opening with locking feature
-Non metal Construction will dent, corrode or conduce electricity
-Water and dust proof allowing the unit to be taken to almost any enviroment
-Case can house up to 2000 meters of fiber
-Auto Purge Valve for changes in altitude and temperature
-Material: SR Polypropylene
-Weight: 0.75kg without fiber
-Operating Temp:-40degree to +55degree
-Typical Loss: 1dB@1310nm for 1000m
-Standard Attachment:Insert Return Loss Report,OTDR Test Report,3D Test Report(Grade A Only)
Place of Origin: Made in China
How to Generate a Baseline Trace Using Fiber Rings
1.Use the Fiber Ring as a launch cable.
Connect the Fiber Ring between your OTDR and the fiber link under test. This will allow you
to measure the loss of the near-end connection.
2.Use the Fiber Ring as a receive cable.
Cconnect the Fiber Ring to the far-end connector of your fiber link under test. This will
allow you measure the loss of the far-end connection.
3.By using Fiber Rings as both launch and receive cables, as shown in the diagram below,
you can measure total insertion loss of the fiber link under test.
OTDR Trace Made Using Launch And Receive Cables.
Example OTDR Test Configuration With Launch And Receive Cables.